- Open Access
A new device (FAQ.FIX®) for orthodontic bracket placement in straight wire technique
Progress in Orthodontics volume 14, Article number: 23 (2013)
In straight wire preadjusted appliances, all the information required to position the teeth in three planes is included in the brackets placed at the midpoint of the facial axis of the clinical crown, defined by facial axis point (FA). Central to this technique is the bracket placement.
Preadjusted orthodontic appliances cannot get the right tooth position with a straight wire because of the inaccuracy of bracket placement. Horizontal, axis, vertical, and base are the most common bracket placement errors.
The aim of this paper was to describe a bracket positioner to fix the brackets accurately (Q) on FA point (FAQ.FIX®) in direct or indirect bonding.
After the development of a prototype, a FAQ.FIX® along with a Bracket Placement Clinical Chart was developed and thus described.
FAQ.FIX® along with the Bracket Placement Clinical Chart may facilitate the accuracy in bracket placement on FA point avoiding the most common bracket placement errors regardless the operator skill, even in particular or difficult case.
FAQ.FIX®may represent a significant improvement in the bracket placement compared to the bracket eye and the traditional gauges positioning. Further studies will be needed to verify the clinical efficacy.
The straight wire technique is the appliance used most widely in orthodontic therapy today. It was introduced by Andrews  in 1972. The basic premise of the preadjusted system is that proper bracket position allows the teeth to be placed with a straight wire into an occlusal contact with an excellent mesiodistal inclination (tip) and excellent faciolingual inclination (torque) .
All the information required to position a tooth in three planes is included in the brackets placed at the midpoint of the facial axis of the clinical crown (FACC), defined by facial axis point (FA). Over the last 40 years, several changes have been made to Andrews' appliance [3, 4] with improvements in preadjusted appliances, without bends on the archwire, to achieve the ideal alignment and leveling, but the most important phase is still the bracket placement .
Poorly positioned brackets result in poorly placed teeth and necessitate more archwire adjustments. This can lead to an increased treatment time or poor occlusion . Several studies [2, 6–9] have demonstrated that the preadjusted orthodontic appliances cannot get the right tooth position with a straight wire because of the poor bracket placement.
Unfortunately, even under the best of circumstances, the ideal bracket placement during initial bonding is often impossible because of the existing malocclusion, operator error, or tooth structure variation [4, 5, 9–12]. Horizontal, axis, vertical, and base are the most common bracket placement errors [2, 4, 6–9]. The ‘eyeball’ bracket position, described by Andrews, cannot be considered reliable and satisfying, as well as, the positioning with the help of a gauge [2, 6–9]. Because of bracket placement errors, orthodontists still spend considerable time in detailing to get the proper alignment of crowns and roots and leveling marginal ridges, particularly near the end of the treatment, to compensate them. Currently, no placement method, direct or indirect, can guarantee the correct execution of this delicate procedure [6, 7, 13–15].
The aim of this paper was to describe a bracket positioner to fix the brackets accurately (Q) on FA point (FAQ.FIX®) in direct or indirect bonding.
In straight wire appliance, the ideal positioner should have the following:
An adjustable locking mechanism for height
References to identify the FACC
An insert for the slot which reproduce the bracket angulation
The torque, instead, included in the base of the bracket, is expressed by base inclination.
The first prototype of the FAQ.FIX® (Cnc Lab Tech, Salerno, Italy) positioner was L-shaped due to horizontal and vertical arms. The insert for the slot reproducing the bracket angulation for each tooth was located on the back and was thick enough to hold the bracket (Figure 1). Developed to place a 5° preangled bracket on an upper left central incisor of a stone model, with a 10-mm FACC, the FAQ.FIX® prototype, without an adjustable locking mechanism for height, had the insert for the slot fixed at 5 mm from the occlusal edge with the same bracket 5° angulation. The FAQ.FIX® prototype worked as tweezers to place the bracket on the selected tooth with the mark on the upper border of the bracket base, and the one on the center of the base of the prototype fell on the FACC (Figure 2). A light-cured composite adhesive was applied to the base.
The FAQ.FIX® prototype test confirmed the accuracy of the bracket placement.
After FAQ.FIX® prototype test, the kit of 20 positioners was produced, one for each tooth and reproducing the preinserted angulation of the single prescription, identified by the international color coding and a distogingival dot (Figure 3). To improve the accuracy and get a unique contact point between the base of the positioner and the tooth, two types of positioners were made, the former for flat edge teeth as the incisors (Figure 4a) and the latter for cuspidate teeth (Figure 4b). Horizontal reference lines were drawn on the vertical arm to set the desired height, whereas vertical reference lines were drawn on the horizontal and vertical arms to identify the FACC.
A screw was positioned on the upper surface to make the mechanism for height adjustable from 3 to 7 mm from the occlusal edge, with 0.5 mm variations (Figures 5, 6 and 7).
The FAQ.FIX® is designed exclusively for straight wire brackets with 0.022 × 0.028 in slot; it is made of polycarbonate for medical use (Bayer Material Science AG, Leverkusen, Germany), and the screw is made of best quality steel. Positioners and screw are made of atoxic material, cold sterilizable and autoclavable at 121° with the ‘plastic instruments’ or ‘delicate’ program for thermosensitive materials.
Each FAQ.FIX® kit (Figure 8) is individually designed for the single prescription and producer. Using FAQ.FIX®, the orthodontist has to follow the instructions carefully, placing it on the tooth, without watching the bracket, and align it to the FACC drawn on the tooth.
In an ideal bracket placement, the FAQ.FIX® positioner allows the match between the center of the insert and the FA point.
Bracket placement clinical chart
The use of a positioning tool in straight wire techniques requires measuring the height of the crown of the teeth, so a ‘Bracket Placement Clinical Chart,’ to keep in the patient's record (Figure 9), has to be developed using the following guidelines:
Measure the height of the crown of each tooth (Figure 10) and report it on the corresponding FACC row
If the same teeth (i.e., cuspids) do not have the same height, due to incomplete eruption or coronal fracture, it may be useful to refer to the completely erupted and/or intact tooth [4, 16–18] (i.e., incisors).
Report the selected FA points for each tooth on FA row
Report the patient overbite to eventually adjust the placement as always done .
For the development of the Bracket Placement Clinical Chart, the operator can also use the MBT or the AACD criteria [4, 16–18], measuring only the height of an upper central incisor to obtain the correct height of the other teeth.
Operative sequence of FAQ.FIX® direct or indirect placement bracket (Figure 11)
Using the Bracket Placement Clinical Chart:
Adjust the height with the screwdriver supplied (Figure 11a).
Introduce the insert into the bracket's slot (Figure 11b).
Apply a light-cured adhesive on the bracket base.
Apply the bracket on the tooth using FAQ.FIX® as tweezers (Figure 11c).
Remove the excess composite.
Indirect bonding bracket placement
The brackets were positioned using FAQ.FIX® on the stone model and then applied accurately on each patient's tooth using a simple and efficient system of indirect bonding that can be easily performed in the orthodontic office using a transfer tray (Figure 12) . An example of indirect bonding bracket placement is reported in Figure 13.
Direct bonding bracket placement
The brackets were positioned using FAQ.FIX® on the teeth, using the same reference used above. The view of the tooth and of FACC line drawn on the tooth, through the ‘window,’ that acts as a viewfinder is easy (Figures 14 and 15).
FACC measurement time of 10 teeth: almost 3 min.
FAQ.FIX® positioners adjustment with 10 brackets insertion: almost 5 min.
Direct/indirect bonding bracket placement: depends by the operator's skill, but according to the authors, it could be faster than the eyeball bracket position or using a gauge.
Written informed consent was obtained from the patient for the publication of this report and any accompanying images.
Results and discussion
The aim of any orthodontist is to place the teeth in an ideal position. In straight wire technique, this should be achieved with preadjusted orthodontic appliance . Unfortunately, an orthodontist may fail this goal even with a preadjusted orthodontic appliance [2, 6–9, 13].
Factors such as bracket placement errors, tooth structure variation [2, 6–10], archwire bending adjustments, and/or rebonding bracket positions are frequently required, and thus, bracket placement remains central to straight wire technique.
The FAQ.FIX® was developed to facilitate the accuracy in bracket placement on FA point.
The FAQ.FIX® along with the Bracket Placement Clinical Chart allows orthodontists to avoid the most common bracket placement errors regardless of operator skill, even in case of partially or fractured teeth or in more difficult cases, such as Class II, Division 2 [11, 12, 20].
The FAQ.FIX®, tested more than 3 years ago and used by the authors and other orthodontists in hundreds of bondings, confirmed the accuracy and validity of systematics.
The authors recommend FAQ.FIX® in the indirect bonding because the bracket placement on a stone model using this positioner is easier and more accurate. The transfer, using transfer trays, on patient's teeth will be fast (almost 15 min including wire insert for self-ligating brackets) and accurate .
Direct bonding bracket placement using FAQ.FIX® may increase the chair time, but it addresses errors of initial bracket positioning decreasing treatment times and obtaining superior results.
The FAQ.FIX® is easy to use, safe, inexpensive, and quick to learn. Moreover, it is a useful teaching tool for young orthodontists, focusing their attention on the bracket placement accuracy.
Correct bracket placement is a problem for both expert and beginning orthodontists. The failure may compromise the treatment outcome. The FAQ.FIX® may represent a significant improvement in bracket placement compared to the eyeball and the traditional gauges bracket position. The authors have verified treatment time reduction using this device because of the initial placement error decrease.
Andrews LF: The six keys to normal occlusion. Am J Orthod 1972, 62: 296–309. 10.1016/S0002-9416(72)90268-0
Balut N, Klapper L, Sandrik J, Bowman D: Variations in bracket placement in the preadjusted orthodontic appliance. Am J Orthod Dentofacial Orthop 1992, 102: 62–67. 10.1016/0889-5406(92)70015-3
Alexander RG, Engel GA: The Alexander Discipline: Contemporary Concepts and Philosophies. Glendora: Ormco Corp; 1986.
McLaughlin RP, Bennet JC: Bracket placement with the preadjusted appliance. J Clin Orthod 1995, 29: 302–311.
Carlson SK, Johnson E: Bracket positioning and resets: five steps to align crowns and roots consistently. Am J Orthod Dentofacial Orthop 2001, 119: 76–80. 10.1067/mod.2001.111220
Koo BC, Chung CH, Vanarsdall RL: Comparison of the accuracy of bracket placement between direct and indirect bonding techniques. Am J Orthod Dentofacial Orthop 1999, 116: 346–351. 10.1016/S0889-5406(99)70248-9
Hodge TM, Dhopatkar AA, Rock WP, Spary DJ: A randomized clinical trial comparing the accuracy of direct versus indirect bracket placement. J Orthod 2004, 31: 132–137. 10.1179/146531204225020427
Armstrong D, Shen G, Petocz P, Darendeliler MA: A comparison of accuracy in bracket positioning between two techniques-localizing the center of the clinical crown and measuring the distance from the incisal edge. Eur J Orthod 2007, 29: 430–436. 10.1093/ejo/cjm037
Miethke RR, Melsen B: Effect of variation in tooth morphology and bracket position on first and third order correction with preadjusted appliances. Am J Orthod Dentofacial Orthop 1999, 116: 329–335. 10.1016/S0889-5406(99)70246-5
Lucchese A, Storti E: Morphological characteristics of primary enamel surfaces versus permanent enamel surfaces: SEM digital analysis. Eur J Paediatr Dent 2011, 12: 179–183.
Perillo L, Cannavale R, Ferro F, Franchi L, Masucci C, Chiodini P, Baccetti T: Meta-analysis of skeletal mandibular changes during Frankel appliance treatment. Eur J Orthod 2011, 33: 84–92. 10.1093/ejo/cjq033
Perillo L, Masucci C, Ferro F, Apicella D, Baccetti T: Prevalence of orthodontic treatment need in southern Italian schoolchildren. Eur J Orthod 2010, 32: 49–53. 10.1093/ejo/cjp050
Joiner M: In-house precision bracket placement with the indirect bonding technique. Am J Orthod Dentofacial Orthop 2010, 137: 850–854. 10.1016/j.ajodo.2009.04.023
Redmond WJ, Redmond MJ, Redmond WR: The OrthoCAD bracket placement solution. Am J Orthod Dentofacial Orthop 2004, 125: 645–646. 10.1016/j.ajodo.2004.01.006
Favero L, Terrazzani C, Favero V, Stellini E, Cocilovo F: Virtual study models: a comparison of modular application systems. Prog Orthod 2009, 10: 16–25.
Lee RL: Esthetics and its relation to function. In Fundamentals of Esthetics, Chapter 5. Edited by: Rufenacht CR. Chicago: Quintessence; 1990:137–209.
Dumont TD OBI Foundation Bioesthetic Dentistry Rev. The ideal biologic dental model 2004, 9.
Hunt K, Turk M: Correlation of AACD accreditation criteria and the Human Biologic Model. J Cosm Dent 2005, 21: 120–31.
Cortesi R, Molinari L: A simple and efficient procedure for indirect bonding. Prog Orthod 2010, 11: 180–184. 10.1016/j.pio.2010.09.004
Farronato G, Maspero C, Farronato D: Orthodontic movement of a dilacerated maxillary incisor in mixed dentition treatment. Dent Traumatol 2009, 25: 451–456. 10.1111/j.1600-9657.2008.00722.x
EM has financial competing interest in FAQ.FIX® because he patented the device. All the other authors declare that they have no competing interests.
FM participated in the sequence alignment and drafted the manuscript. EM patented the device, conceived of the study, and participated in its design and coordination. VA participated in the sequence alignment. JS participated in the design of the study and performed the statistical analysis. LP conceived of the study and participated in its design and coordination. All authors read and approved the final manuscript.
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
Rights and permissions
Open Access This article is distributed under the terms of the Creative Commons Attribution 2.0 International License (https://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
About this article
Cite this article
Mazzeo, F., Marchese, E., Assumma, V. et al. A new device (FAQ.FIX®) for orthodontic bracket placement in straight wire technique. Prog Orthod. 14, 23 (2013). https://doi.org/10.1186/2196-1042-14-23
- Tooth Position
- Stone Model
- Straight Wire
- Bracket Base
- Single Prescription