Mini-implants and the efficiency of Herbst treatment: a preliminary study
© Luzi et al.; licensee Springer. 2013
Received: 3 July 2012
Accepted: 22 April 2013
Published: 31 July 2013
The purpose of this study is to present the use of a modified Herbst appliance in association with temporary anchorage devices (TADs) in order to enhance the correction of skeletal class II malocclusions.
Ten consecutive adolescents scheduled for Herbst treatment were assigned to two treatment groups. Five cases were treated with a modified miniscrew-supported Herbst appliance (experimental group (EG)) and five cases with a conventional cast Herbst appliance (control group (CG)). In all cases, the Herbst was kept in place for 9 months and was followed by fixed appliances until class I relationships were achieved. The initial (T1) and final (T2) lower incisor inclination on lateral headfilms were analyzed for each case, and the mean increase for the five EG patients and the five CG patients were compared.
The mean increase in lower incisor inclination at the end of treatment was 1° (range 0° to 2°) for the EG and 7° (range 4° to 10°) for the CG.
The rational association of TADs with the Herbst appliance can optimize treatment efficiency and skeletal response by reducing the occurrence of excessive lower incisor proclination.
The introduction of noncompliance systems has markedly influenced class II treatment. Devices with the purpose of distalizing upper molars[2–7] as well as devices with the purpose of advancing the mandible[8–13] have become very popular in the last two decades. However, in spite of the popularity of these devices, they all lead to some anchorage loss and undesirable side effects as the anchorage teeth do never exhibit complete stability. Herbst treatment, in particular, generally features an unwanted proclination of the lower anterior segment[14–16]. The introduction of temporary anchorage devices (TADs) has recently brought the option of absolute anchorage control to daily clinical practice. While this benefit has been extended to distalizing devices[17–20], eliminating anterior anchorage loss, and generating new and more efficient treatment protocols, the association of bite-jumping devices and TADs has been discussed but not yet evaluated in a controlled clinical study. The aim of this study, therefore, is to evaluate whether the association of a modified Herbst appliance and TADs can enhance the correction of skeletal class II malocclusion, avoiding the undesirable proclination of the lower anterior teeth.
Values of initial and final lower incisor inclinations and ANB angles of all the patients
Initial ANB angle
Final ANB angle
Results and discussion
This preliminary study describes the association of a modified Herbst appliance and TADs comparing the effects of this new protocol to traditional Herbst treatment. To date, with a history of over 100 years, the Herbst appliance is still considered a state-of-the-art solution and the most used noncompliance device for class II treatment[23–26]. All fixed bite-jumping devices generate both skeletal and dental effects, resulting as a main side effect in a proclination of the lower incisors, contributing to overjet reduction[14–16]. This undesired effect can minimize the skeletal effects and impede the desired forward displacement of the mandible, thereby preventing the establishment of a solid class I final dental relationship and a tight intercuspation. Although many attempts have been made to avoid lower incisor proclination, e.g., with the use of cast mandibular appliances, archwires with torque bends, or brackets with selective torques, absolute anchorage control has not yet been achieved and loss of mandibular anchorage can be anticipated[27, 28]. This is especially undesirable in patients where the lower incisors already exhibit a compensatory proclination at the start of treatment, for which reason, further proclination should be avoided during the correction of the sagittal discrepancy. This was taken into consideration when the patients were assigned to the experimental or the control group. A randomized assignment to groups was considered unethical, and the five cases with the greatest proclination of the lower incisors measured to the mandibular plane were therefore assigned to the EG. To reduce bias, the same bracket prescription was used in all the cases, and patients either requiring extractions or with lower arch crowding greater than 4 mm were excluded from the sample. Therefore, the changes in lower incisor position that resulted from leveling and alignment or the space closure phases were excluded. Most cephalometric errors are either due to experience of the operator or to landmark identification. All landmark identification and measurements were carried out twice by the same experienced operator. In case of discrepancy between the first and second measurements, a third measurement was then performed.
Although TADs have become a daily clinical tool in orthodontics, their use, in combination with the Herbst appliance, has not yet been standardized into a treatment protocol or reported in clinical trials. This preliminary study describes how a combination of miniscrews and Herbst appliance can control the proclination of the lower incisors during treatment. The mandibular alveolar process has optimal cortical bone thickness and adequate inter-root distance for insertion of miniscrews with standard characteristics (1.5-mm diameter and 6-mm thread length)[30, 31]. Following the insertion and in the presence of primary stability, a stiff connection to the Herbst appliance was required. This was achieved by means of a 0.12-mm stainless steel ligature wire. In this way, the Herbst was directly connected to the lower dentition and indirectly to the lower basal bone with the aim of enhancing the skeletal effect and reducing the dentoalveolar compensations in the mandibular arch. Direct orthopedic load on miniscrews should be avoided as it most likely would increase the risk of failure of these devices designed to withstand standard orthodontic forces. In all cases, the Herbst appliance was kept in place for a minimum of 9 months (mean duration of Herbst phase 9.6 months) to induce adequate skeletal and neuromuscular adaptation. The stability of the Herbst in both groups and the stability of the TADs and tightness of the ligature wires in the EG were routinely checked monthly. The two miniscrews that lost stability were either positioned too close to a dental root or loosened due to adverse local conditions (poor bone quality, inflammation, etc.). It is well known that mini-implants experience failure, and various success rates are reported without significant differences between immediately loaded or delayed loaded samples. The conditions of the load (immediate and indirect) could not be blamed for the loss of stability. The success rates of the insertion sites, between the mandibular premolars and the second premolar and first molar, have not shown to differ from other mandibular or maxillary sites in growing patients. Patient gender and side of insertion can also be considered irrelevant in relationship to success rates. Although the mandibular inter-radicular space between the first and second premolars is, on average, greater than the one between the second premolar and first molar, individual variations should be considered and periapical radiographs should always be used for implant site determination.
At the end of treatment, all cases displayed class I canine and molar relationships and reduction of the ANB angle. The results thus confirm the efficiency of the Herbst appliance in the correction of class II malocclusions. This corroborates the findings of Schaefer et al. who reported normalization of the dentoskeletal parameters following overall treatment in their stainless steel crown Herbst sample. However, there was a clear difference of the change in lower incisor inclination between the two groups studied. The side effects were reduced in the EG (Table 1). While the mean increase in the inclination of the lower incisors from T1 to T2 in the CG is comparable to the one reported in other similar studies with greater sample size[14, 39], as a result of the absolute anchorage delivered by the TADs, this proclination was smaller in the EG. In addition, the mean reduction in ANB angle was greater in the EG (Table 1), confirming the fact that reducing the dentoalveolar compensations during Herbst treatment can help to achieve a greater skeletal effect. Although the EG showed reduced dentoalveolar compensations compared to the CG, the results should be interpreted with caution due to the small sample size.
The association of TADs with the Herbst appliance can be applied in order to reduce undesirable proclination of lower incisors and enhance the skeletal response. The simple use of mini-implants makes it possible for the orthodontist to develop a new ideal treatment protocol facilitating the correction of class II malocclusions in those adolescents requiring mandibular advancement with minimal dentoalveolar compensations. Further studies and clinical trials with a larger sample size are recommended to confirm the results of this preliminary study.
Written informed consent was obtained from the patients for publication of this report and any accompanying images.
- Buckhardt D, McNamara J, Baccetti T: Maxillary molar distalization or mandibular enhancement: a cephalometric comparison of comprehensive orthodontic treatment including the Pendulum and the Herbst appliances. Am J Orthod Dentofacial Orthop. 2003, 123: 108–16. 10.1067/mod.2003.7View ArticleGoogle Scholar
- Hilgers JJ: The pendulum appliance for class II noncompliance therapy. J Clin Orthod. 1992, 26: 706–14.PubMedGoogle Scholar
- Jones RD, White MJ: Rapid class II molar correction with an open coil jig. J Clin Orthod. 1992, 26: 661–4.PubMedGoogle Scholar
- Locatelli R, Bednar J, Dietz VS, Giannelly AA: Molar distalization with super-elastic NiTi wire. J Clin Orthod. 1992, 26: 277–9.PubMedGoogle Scholar
- Carano A, Testa M: The distal jet for upper molar distalization. J Clin Orthod. 1996, 30: 374–80.PubMedGoogle Scholar
- Caprioglio A, Beretta M, Lanteri C: Maxillary molar distalization: pendulum and fast-back, comparison between two approaches for class II malocclusion. Prog Orthod. 2011, 12: 8–16. 10.1016/j.pio.2011.02.007View ArticlePubMedGoogle Scholar
- Fortini A, Lupoli M, Parri M: The first class appliance for rapid molar distalization. J Clin Orthod. 1999, 33: 322–8.PubMedGoogle Scholar
- Jasper JJ: Jasper Jumper: a fixed functional appliance. Sheboygan: American Orthodontics; 1987.Google Scholar
- De Vincenzo JP: The Eureka spring: a new interarch force delivery system. J Clin Orthod. 1997, 31: 454–67.Google Scholar
- Eckhart JE: Introducing the MARA. Clin Impressions 1998,7(2–5):24–7.Google Scholar
- Awbrey JJ: The bite fixer. Clin Impressions. 1999, 8: 10–7.Google Scholar
- Heinig N, Goz G: Clinical application and effects of the Forsus spring: a study of a new Herbst hybrid. J Orofac Orthop. 2001, 62: 436–50. 10.1007/s00056-001-0053-6View ArticlePubMedGoogle Scholar
- Corbett MC, Molina FG: Twin force bite corrector: fight force and patient friendly. Syllabus, Carlsbad: Ortho Organizers; 2001.Google Scholar
- Martin J, Pancherz H: Mandibular incisor position changes in relation to amount of bite jumping during Herbst/multibracket appliance treatment: a radiographic-cephalometric study. Am J Orthod Dentofacial Orthop. 2009, 136: 44–51. 10.1016/j.ajodo.2007.07.027View ArticlePubMedGoogle Scholar
- Barnett GA, Higgins DW, Major PW, Flores-Mir C: Immediate skeletal and dentoalveolar effects of the crown- or banded type Herbst appliance on class II division I malocclusion. Angle Orthod. 2008, 78: 361–9. 10.2319/031107-123.1View ArticlePubMedGoogle Scholar
- Weschler D, Pancherz H: Efficiency of three mandibular anchorage forms in Herbst treatment: a cephalometric investigation. Angle Orthod. 2004, 75: 23–7.Google Scholar
- Wehrbein H, Glatzmaier J, Mundwiller U, Diedrich P: The orthosystem - a new system for orthodontic anchorage in the palate. J Orofac Orthop. 1996, 57: 142–53. 10.1007/BF02191878View ArticlePubMedGoogle Scholar
- Velo S, Rotunno E, Cozzani M: The implant distal jet. J Clin Orthod. 2007, 41: 88–93.PubMedGoogle Scholar
- Kinzinger G, Gülden N, Yildizhan F, Diedrich P: Efficiency of a skeletonized distal jet appliance supported by miniscrew anchorage for noncompliance maxillary molar distalization. Am J Orthod Dentofacial Orthop. 2009, 136: 578–86. 10.1016/j.ajodo.2007.10.049View ArticlePubMedGoogle Scholar
- Ludwig B, Glasl B, Kinzinger GS, Walde KC, Lisson JA: The skeletal frog appliance for maxillary molar distalization. J Clin Orthod. 2011, 45: 77–84.PubMedGoogle Scholar
- Al-Kalaly AA, Wong RW, Cheung LK, Purkayastha SK, Schätzle M, Rabie AB: Evaluation of bone thickness around the mental foramen for potential fixation of a bone-borne functional appliance: a computer tomography scan study. Clin Oral Impl Res. 2010, 21: 1288–93. 10.1111/j.1600-0501.2010.01947.xView ArticleGoogle Scholar
- Luzi C, Luzi V: Skeletal class II treatment with the miniscrew-anchored Herbst. Orthodontie Française 2013. in press in pressGoogle Scholar
- Herbst E: Atlas und Grundriss der Zahnärztllichen Orthopädie. Munich: JF Lehmann Verlag; 1910.Google Scholar
- Pancherz H: The Herbst appliance—it’s biologic effects and clinical use. Am J Orthod. 1985, 87: 1–20. 10.1016/0002-9416(85)90169-1View ArticlePubMedGoogle Scholar
- Flores-Mir C, Ayeh A, Goswani A, Charkhandeh S: Skeletal and dental changes in class II division I malocclusions treated with splint-type Herbst appliances. Angle Orthod. 2007, 77: 376–81. 10.2319/0003-3219(2007)077[0376:SADCIC]2.0.CO;2View ArticlePubMedGoogle Scholar
- Konik M, Pancherz H, Hansen K: The mechanism of class II correction in late Herbst treatment. Am J Orthod Dentofacial Orthop. 1997, 112: 87–91. 10.1016/S0889-5406(97)70278-6View ArticlePubMedGoogle Scholar
- Pancherz H: The effects, limitations, and long-term dentofacial adaptations to treatment with the Herbst appliance. Semin Orthod. 1997, 3: 232–43. 10.1016/S1073-8746(97)80056-4View ArticlePubMedGoogle Scholar
- Purkayastha SK, Rabie AB, Wong R: Treatment of skeletal class II malocclusion in adults: stepwise vs single-step advancement with the Herbst appliance. World J Orthod. 2008, 9: 233–43.PubMedGoogle Scholar
- Broch J, Slagsvold O, Røsler M: Error in landmark identification in lateral radiographic headplates. Eur J Orthod. 1981, 3: 9–13. 10.1093/ejo/3.1.9View ArticlePubMedGoogle Scholar
- Lee KJ, Joo E, Kim KD, Lee JS, Park YC, Yu HS: Computed tomographic analysis of tooth-bearing alveolar bone for orthodontic miniscrew placement. Am J Orthod Dentofacial Orthop. 2009, 135: 486–94. 10.1016/j.ajodo.2007.05.019View ArticlePubMedGoogle Scholar
- Farnsworth D, Rossouw PE, Ceen RF, Buschang P: Cortical bone thickness at common mini-implant placement sites. Am J Orthod Dentofacial Orthop. 2011, 139: 495–503. 10.1016/j.ajodo.2009.03.057View ArticlePubMedGoogle Scholar
- Luzi C, Verna C, Melsen B: Guidelines for success in placement of orthodontic mini-implants. J Clin Orthod. 2009, 43: 39–44.PubMedGoogle Scholar
- Pancherz H: The modern Herbst appliance. In Dentofacial orthopedics with functional appliances. 2nd edition. Edited by: Graber TM, Rakosi T, Petrovic AG. St Louis: Mosby; 1997:336–66.Google Scholar
- Kuroda S, Yamada K, Deguchi T, Hashimoto T, Kyung HM, Takano-Yamamoto T: Root proximity is a major factor for screw failure in orthodontic anchorage. Am J Orthod Dentofacial Orthop. 2007, 131: S68–73. 10.1016/j.ajodo.2006.06.017View ArticlePubMedGoogle Scholar
- Schätzle M, Männchen R, Zwahlen M, Lang NP: Survival and failure rates of orthodontic temporary anchorage devices: a systematic review. Clin Oral Implants Res. 2009, 20: 1351–9. 10.1111/j.1600-0501.2009.01754.xView ArticlePubMedGoogle Scholar
- Moon CH, Lee DG, Lee HS, Im JS, Baek SH: Factors associated with the success rate of orthodontic miniscrews placed in the upper and lower posterior buccal region. Angle Orthod. 2008, 78: 101–6. 10.2319/121706-515.1View ArticlePubMedGoogle Scholar
- Poggio PM, Incorvati C, Velo S, Carano A: “Safe zones”: a guide for miniscrew positioning in the maxillary and mandibular arch. Angle Orthod. 2006, 76: 191–7.PubMedGoogle Scholar
- Cozza P, Baccetti T, Franchi L, De Toffol L, McNamara JA Jr: Mandibular changes produced by functional appliances in class II malocclusion: a systematic review. Am J Orthod Dentofacial Orthop 2006, 129: 599. e1–e12 e1–e12PubMedGoogle Scholar
- Schaefer AT, McNamara JA Jr, Franchi L, Baccetti T: A cephalometric comparison of treatment with the Twin-block and stainless steel crown Herbst appliances followed by fixed appliance therapy. Am J Orthod Dentofacial Orthop. 2004, 126: 7–15. 10.1016/j.ajodo.2003.06.017View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.