- Open Access
A comparison of three-dimensional stress distribution and displacement of naso-maxillary complex on application of forces using quad-helix and nickel titanium palatal expander 2 (NPE2): a FEM study
© The Author(s). 2016
- Received: 4 March 2016
- Accepted: 19 May 2016
- Published: 8 June 2016
Our objectives are to analyse and to compare the stress distribution and displacement of the craniofacial structures, following the application of forces from quad-helix and Nickel Titanium Palatal Expander-2 (NPE2) using finite element analysis.
Three-dimensional finite element models of young dried human skull, quad-helix appliance and NPE2 were constructed, and the initial activation of the expanders was stimulated to carry out the analysis and to evaluate the Von Misses stresses and displacement.
Both the models demonstrated the highest stresses at the mid-palatal suture, with maximum posterior dislocation. The second highest stress was recorded at the fronto-zygomatic suture. The pattern of stress distribution was almost similar in both the groups, but NPE2 revealed lower magnitude stresses than quad-helix. The only exception being quad-helix model showed high stress levels around pterygo-maxillary suture whereas minimal stress around pterygo-maxillary suture was noticed after NPE2 activation. The cusp of the erupting canine and the erupting mesiobuccal cusp of the second molar showed outward, backward and downward displacement signifying increase in their eruption pattern following maxillary expansion.
Maxillary expansion using quad-helix and NPE2 can be used in posterior crossbite correction in cases where maximum skeletal changes are desirable at a younger age; it is furthermore effective in treating young patients with impacted or displaced teeth. Quad-helix and NPE2 produced acceptable forces for orthopaedic treatment even after being orthodontic appliances; their clinical application should be correctly planned as the effects of these appliances are largely age dependent.
- Orthodontic Appliance
- Nickel Titanium
- Mixed Dentition
- Rapid Maxillary Expansion
- Lateral Nasal Wall
Maxillary expansion treatments have been used for more than a century to correct maxillary transverse deficiency. The earliest common cited report was that of E.C. Angell published in Dental cosmos in 1860; however, the work was discredited at that time but the technique was generally accepted . Slow maxillary expansion produces more physiologic response at the mid-palatal suture area. It produces less tissue resistance, since it delivers constant physiologic force in the suture and allows better bone formation, both these factors help to minimize the post expansion relapse .
Quad-helix has been evolved from the W arch of coffin by the incorporation of helical loops, two anteriorly and two posteriorly, which increased the flexibility and range of action of the appliance . Arndt  introduced nickel titanium palatal expanders in the year 1993. Corbett  introduced Nickel Titanium Palatal Expander-2 (NPE2) in 1997. It generates optimal and constant pressure which is a consequence of nickel titanium’s shape memory and effects of transition temperature. It delivers a uniform, slow, continuous force for maxillary expansion, molar rotation, distalization and arch development.
Many research studies had been done in comparison between slow maxillary expansion and rapid maxillary expansion techniques, but very little had been done to compare differences between various techniques of slow maxillary expansion. Finite element analysis is an engineering method of calculating stresses and strains in materials, including living tissues . Finite element method is closer to the clinical situation, and we accept that the qualitative behaviour of a dry skull does not simulate with a high degree of accuracy in the clinical situation, it is possible to indicate the way that the two maxillary halves separate during expansion and how the effect of the force influences the other craniofacial structures.
The main aim of this study was to analyse stress distribution and displacement of the craniofacial structures on the application of forces induced by quad-helix and NPE2 using finite element analysis.
For creating a finite element model, a computer-aided design model was constructed from a dry young human skull of an approximate age of 12 years. Age estimation was carried out by dental eruption pattern. Human maxillary skull bone without mandible was checked for defects or discontinuity in the craniofacial anatomy. CT scan images of the maxillary bone were taken by SIEMENS SOMATOM Definition 64 (kVp120; mAs 290) in axial direction. Sequential CT images were taken at 0.5-mm intervals to reproduce finer and detailed aspects of the geometry. This scheme of model creation was intended to improve the anatomical accuracy over the previous methodologies where CT sections were taken at 1, 5, and 10 mm, respectively.
ANSYS Professional NLS (Version 16; ANSYS Inc.) was used to carry out the analysis. The total numbers of elements in the geometry were 864,650, and total numbers of nodes created were 247,119.
Thickness of the cortical bone was determined according to the study by Farnsworth et al. , thickness of periodontal ligament was 0.2 mm , and the thickness of the maxillofacial and mid-palatal sutures were 0.5 mm .
Laying boundary conditions
The nodes of the mid-palatal suture element that were created in this study were placed on the symmetrical plane and were left unconstrained. Nodes along the foramen magnum and on the centre of the forehead were constrained in all degrees of freedom, with zero displacement and zero rotation . The 3D coordinates were X, transverse plane; Y, sagittal plane; and Z, vertical plane. The midpoint of the lingual alveolar ridge of each tooth was used as a reference point to evaluate alveolar bone displacement. Positive values indicate outward, backward, and downward displacement on the X, Y, and Z planes, respectively.
In this study, a force of 350 g is used in case of NPE2 model as a 3-mm increment of expansion with NiTi palatal expander produces 350 g of force . Quad-helix expansion appliance was activated to one molar width, i.e., around 8 mm. Chaconas  reported that an initial 8 mm expansion of quad-helix prior to cementation created approximately 14 oz (396.89 g) of force which is equivalent to 3.89 N, and therefore, 2 N of force was applied on each side. The displacements of maxillofacial complex and Von Misses stresses in different parts were studied. Von Misses is a criteria used in predicting the onset of yield in ductile materials.
Displacements of various structures in all the three planes produced after the activation of both the appliances
Von Misses stress distribution over the Naso-maxillary complex after the activation of both the appliances
Maxillofacial landmarks displacement (mm) after activation of quad-helix and NPE2
Frontal process of zygomatic bone
Maxillary process of zygomatic bone
Temporal process of zygomatic bone
Lateral nasal wall
Inferior nasal floor
Lateral pterygoid plate
Medial pterygoid plate
Mid-palatal suture displacement (mm) after activation of quad-helix and NPE2
Dento-alveolar displacement (mm) after activation of quad-helix and NPE2
Lingual marginal ridge of central incisor
Lingual marginal ridge of lateral incisor
Lingual marginal ridge of first permanent molar
Central incisor cusp tip
First permanent molar mesiobuccal cusp tip
Erupting canine cusp tip
Erupting second molar mesiobuccal cusp tip
Stress distribution (MPa) at the maxillofacial sutures and landmarks after activation of the quad-helix and NPE2 appliance
Frontal process of zygomatic bone
Maxillary process of zygomatic bone
Temporal process of zygomatic bone
Lateral nasal wall
Inferior nasal floor
Lateral pterygoid plate
In young patients, slow maxillary expansion is said to provide the maximum rate at which the mid-face sutures can adapt, with minimum tearing and haemorrhaging compared with rapid maxillary expansion [14–16]. Animal and histological studies indicate that slow maxillary expansion improves conservation of the suture and can produce a more stable result than rapid maxillary expansion [14, 15] Some clinical studies also suggest that slow maxillary expansion is more stable than rapid maxillary expansion . Changes in craniofacial skeleton arising from orthodontic treatment are more complex than envisaged from two-dimensional cephalometric assessments, and so we decided to do a three-dimensional study of the craniofacial skeleton using finite element analysis.
In the present study, we witnessed skeletal changes following slow maxillary expansion, similar to those Hicks  had reported; according to him, substantial skeletal changes with slow maxillary expansion can be observed especially in younger children. The theory is that the main resistance to the opening of the mid-palatal suture is not the suture itself but the surrounding tissues such as the circum-maxillary structures and mid-face sutures . This observation lends support to studies that noted the buttressing effect of the zygomatic processes against forces of expansion  supporting to evidence what we noticed in our study.
Orthopaedic force distribution after the activation of both the appliances quad-helix and NPE2 were observed to be analogous to what Chaconas and Caputo  stated that there was the buttressing of the maxillary tuberosity with the pterygoid plates; the sphenoid bone allowed the forces to then radiate to the base of the medial pterygoid plate from this region the forces then branched superiorly toward the malar and zygomatic bones. Specifically, the areas of the zygomatico-maxillary and zygomatico-temporal sutures were affected. The forces then radiated supero-medially toward the medial wall of the orbit and concentrated at the junction of the nasal and lachrymal bones. From our study, it is evident that the maxillary buttresses are the main areas of resistance with the forces on the maxillary molars; the stress radiates to the three main buttresses of the mid-face cranial complex: the naso-maxillary, the zygomatico-maxillary, and the pterygo-maxillary .
Sandikcioglu et al.  in his study achieved more posterior expansion of the palate; however, both the models in the present study exhibited similar results showing greater posterior dislocation of the mid-palatal suture than in the anterior region.
Our study exhibits downward and forward displacement of maxilla similar to the displacement observed by Jafari et al.  who studied stress distribution and displacement of various craniofacial structures following transverse orthopaedic forces. In the present study, we found backward movement of the point A which can be supported by Wertz  and Sandikcioglu et al.  who reported that the point A moved slightly backward and the ANB mostly showed high values. According to Wertz research, if it is assumed that point A does not move forward during rapid maxillary expansion, the change in the ANB angle could be a result of posterior rotation of point B .
During expansion, not all changes are caused by alveolar bending but are partly due to the tipping of teeth in the alveolar bone; this tipping is usually accompanied by some extrusion . Similar outcomes were seen in the present study where tipping and slight extrusion of the molars were seen. Herold  presented greater buccal tipping in a sample treated with quad-helix. Shetty et al.  demonstrated tipping and extrusion of teeth following the use of NPE2.
It is witnessed in the present study that when correctly employed, the quad-helix can produce results similar to the rapid maxillary expansion and also correct all the transverse problems in the growing patients . In the same way, NPE2 showed orthopaedic changes when used in mixed dentition, which is reinforced by the findings of studies that NPE2 even though being an orthodontic appliance showed orthopaedic changes .
The cusp of the erupting canine and the mesiobuccal cusp of the erupting second molar showed outward, backward and downward displacement indicating the speeding up for eruption which was similar to study  where rapid maxillary expansion was effective in treating patients in the late mixed dentition with palatally displaced canines. Baccetti et al.  stated that maxillary expansion is effective as an interceptive procedure to prevent final impaction of maxillary canines with palatal displacement in the early mixed dentition.
Quad-helix model showed high stress levels around pterygo-maxillary suture, whereas the stress levels were very minimal in NPE2 model which supports the findings of Donohue et al.  who identified that both the quad-helix and NPE2 were equally efficacious maxillary expanders; however, according to them, quad-helix appliance produced more controlled differential expansion between the first molars than NPE2 in their clinical comparison between the two and so they stated quad-helix more individually predictable in expansion.
The results of the present study using three-dimensional finite element model of a young skull provided explanation about the response of appliance activation within the bony tissues; however, finite element method has certain limitations like the results being applicable to the generated model and may not necessary apply to all individuals; moreover, clinical environment cannot be created in the model like mastication forces and patient movements; therefore, the results are for qualitative purpose for emphasizing skeletal responses of the appliances in human tissues.
Quad-helix and NPE2 produced acceptable forces for orthopaedic treatment even after being an orthodontic appliance; their clinical application should be correctly planned as the effects of these appliances are largely age dependent. Both of these appliances can be used alternatively in posterior crossbite, where skeletal changes are desired at a younger age. Quad-helix showed more skeletal expansion where as NPE2 is proved to be more advantageous when dental expansion is desired. Maxillary expansion is furthermore effective in treating patients with impacted or displaced teeth because of arch length insufficiency during mixed dentition.
Although both the appliances produced almost similar amounts of maxillary expansion, from our study, we conclude that the quad-helix treatment regime is considered to be more successful if orthopaedic results are anticipated during mixed dentition period.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Timms DJ. The dawn of rapid maxillary expansion. Angle Orthod. 1999;69:247–51.PubMedGoogle Scholar
- Barber AF, Sims MR. Rapid maxillary expansion and external root resorption in man: a scanning electron microscope study. Am J Orthod. 1981;79:630–52.View ArticlePubMedGoogle Scholar
- Chaconas SJ, de Alba y Levy JA. Orthopedic and orthodontic applications of the quad-helix appliance. Am J Orthod. 1977;72:422–8.View ArticlePubMedGoogle Scholar
- Arndt WV. Nickel titanium palatal expander. J Clin Orthod. 1993;27:129–37.PubMedGoogle Scholar
- Corbett MC. Slow and continuous maxillary expansion, molar rotation and molar distalization. J Clin Orthod. 1997;4:253–63.Google Scholar
- Srirekha A, Bashetty K. Infinite to finite: an overview of finite element analysis. Indian J Dent Res. 2010;21:425–32.View ArticlePubMedGoogle Scholar
- Farnsworth D, Rossouw PE, Ceen RF, Buschang PH. Cortical bone thickness at common mini-screw implant placement sites. Am J Orthod Dentofac Orthop. 2011;139:495–503.View ArticleGoogle Scholar
- Kronfeld R. Histological study of the influence of function on the human periodontal membrane. J Am Dent Assoc. 1931;18:1242–74.Google Scholar
- Fricke-Zech S, Gruber RM, DullinC ZA, Kramer FJ, Meesenburg KD, et al. Measurement of the mid-palatal suture width. Angle Orthod. 2012;82:145–50.View ArticlePubMedGoogle Scholar
- Lee NK, Beak SH. Stress and displacement between maxillary protraction with miniplates placed at the infrazygomatic crest and the lateral nasal wall: a 3-dimensional finite element analysis. Am J Orthod Dentofac Orthop. 2012;141:345–51.View ArticleGoogle Scholar
- Tanaka OM, Saga AY, Pithon MM, Argenta MA. Stresses in the mid-palatal suture in the maxillary protraction therapy: a 3D finite element analysis. Prog Orthod. 2016;17:8.View ArticlePubMedPubMed CentralGoogle Scholar
- Naceur IB, Charfi A, Bouraoui T, Elleuch K. Finite element modelling of superelastic nickel-titanium orthodontic wires. J Biomechanics. 2014;47:3630–8.View ArticleGoogle Scholar
- Moon W, Wu KW, MacGuiness M, Sung J, Youssef G, Machado A. The efficacy of maxillary protraction protocols with the micro-implant-assisted rapid palatal expander (MARPE) and the novel N2 mini-implant—a finite element study. Prog Orthod. 2015;16:16.View ArticlePubMedPubMed CentralGoogle Scholar
- Bell RA. The effects of maxillary expansion using a quad-helix appliance during the deciduous and mixed dentitions. Am J Orthod. 1981;79:152–61.View ArticlePubMedGoogle Scholar
- Storey E. Tissue response to the movement of bones. Am J Orthod. 1973;64:229–47.View ArticlePubMedGoogle Scholar
- Mossaz-Joëlson K, Mossaz CF. Slow maxillary expansion: a comparison between banded and bonded appliances. Eur J Orthod. 1989;11:67–76.PubMedGoogle Scholar
- Shetty V, Caridad JM, Caputo AA, Chaconas SJ. Biomechanical rationale for surgical-orthodontic expansion of the adult maxilla. J Oral Maxillofac Surg. 1994;52:742–9.View ArticlePubMedGoogle Scholar
- Ladner PT, Muhl ZF. Changes concurrent with orthodontic treatment when maxillary expansion is a primary goal. Am J Orthod. 1995;108:184–93.View ArticleGoogle Scholar
- Chaconas SJ, Caputo AA. Observation of orthopedic force distribution produced by maxillary orthodontic appliances. Am J Orthod. 1982;82:492–501.View ArticlePubMedGoogle Scholar
- MacGinnis M, Chu H, Youssef G, Wu KW, Machado AW, Moon W. The effects of micro-implant assisted rapid palatal expansion (MARPE) on the nasomaxillary complex: a finite element method (FEM) analysis. Prog Orthod. 2014;15:52.View ArticlePubMedPubMed CentralGoogle Scholar
- Sandikcioglu M, Hazar S. Skeletal and dental changes after maxillary expansion in the mixed dentition. Am J Orthod Dentofac Orthop. 1977;111:321–7.View ArticleGoogle Scholar
- Jafari A, Shetty KS, Kumar M. Study of stress distribution and displacement of various craniofacial structures following application of transverse orthopedic forces: a three-dimensional FEM study. Angle Orthod. 2003;73:12–21.PubMedGoogle Scholar
- Wertz RA. Skeletal and dental changes accompanying rapid mid-palatal suture opening. Am J Orthod. 1970;58:44–66.View ArticleGoogle Scholar
- Hicks EP. Slow maxillary expansion. A clinical study of the skeletal versus dental response to low-magnitude force. Am J Orthod. 1978;73:121–41.View ArticlePubMedGoogle Scholar
- Herold JS. Maxillary expansion: a retrospective study of three methods of expansion and their long term sequelae. Br J Orthod. 1989;16:195–200.View ArticlePubMedGoogle Scholar
- Shetty P, Hegde AM, Rai K. Study of stress distribution and displacement of the maxillary complex following application of forces using jackscrew and nitanium palatal expander 2-A Finite element study. J Clin Paediatric Dent. 2009;34:87–93.View ArticleGoogle Scholar
- Sigler LM, Baccetti T, McNamara Jr JA. Effect of rapid maxillary expansion and transpalatal arch treatment associated with deciduous canine extraction on the eruption of palatally displaced canines: a 2-center prospective study. Am J Orthod Dentofac Orthop. 2011;139:e235–44.View ArticleGoogle Scholar
- Baccetti T, Mucedero M, Leonardi M, Cozza P. Interceptive treatment of palatal impaction of maxillary canines with rapid maxillary expansion: a randomized clinical trial. Am J Orthod Dentofac Orthop. 2009;136:657–61.View ArticleGoogle Scholar
- Donohue VE, Marshman LAG, Winchester LJ. A clinical comparison of the Quadhelix appliance and the nickel titanium (tandem loop) palatal expander: a prelimnary prospective investigation. Eur J Orthod. 2004;26:411–20.View ArticlePubMedGoogle Scholar